Real-Life RLC circuit - why so lopsided? I am using a superconducting $RLC$-circuit for detecting weak electrical signals. The center-frequency is around 650 kHz, and the $Q$-value approximately 50k. I am trying to better understand my signal background. Without the signal source, the electrical schematic be simplified to this: 

Note that there is no input to this circuit: We are looking at the Johnson-noise of the detection circuit in combination with the noise-floor of the amplifier's input stage.
Ideally, the power spectrum of the Johnson noise is given by $|Z_{RLC}|^2$, which can be shown to be a Lorentzian:
$$ \text{noise in dB} = 10 \log_{10}\left( 10^{A/10} \frac{\left(\frac{\nu_0}{2 Q}\right)^2}{\left(\nu-\nu_0\right)^2 + \left(\frac{\nu_0}{2 Q}\right)^2} + 10^{b/10}\right) \quad .$$
In this form, the amplitude $A$ and the background $b$ are given in dB. The center of the resonance is given by $\nu_0$, and the width given by the $Q$-value with $Q = \nu_0/\text{FWHM}$. 
In reality, the resonance is almost a Lorentzian, but is lopsided. This becomes especially obvious when using a Lorentzian as a fit-function:

I checked a bit further away from the resonance to make sure that the transfer function of my amplifier is reasonably flat. Therefore, I don't think it can explain the lopsidedness. I suspect the lopsidedness is caused by an additional resonance, for example some by parasitic $R$, $L$ or $C$ somewhere in the system, like the series/parallel resonances in a Quartz oscillator.  But I cannot find a reasonable model for that. What kind of parasitic component could cause this lopsidedness?
Possibly relevant: The amplifier is not directly connected to the "hot end" of the resonator, but to a 1:10 tap close to the "cold end" (ground). 
 A: If you measured this with a frequency sweep excitation, There will be some lopsidedness if the sweep rate is too high relative to the Q of the circuit. The "undershoot and rebound" in amplitude above the resonance is what sweeping too fast usually looks like, assuming the sweep was increasing the frequency. If the sweep rate is much too fast, you can get several "bounces" in the response before it settles onto the true response of the system.
ISO-7626 gives standard values for sweep rate for mechanical vibration testing. The same principle applies to electronic circuits. 
A typical order of magnitude for the maximum sweep rate to get acceptable results is $f^2/Q^2$  Hz/sec where $f$ is the resonance frequency in Hz. A simple check is to compare two frequency sweeps with the frequency increasing and decreasing through the resonance.
For mechanical vibration with low frequency resonances (e.g. 10 - 100 Hz) and high Q values, an accurate frequency sweep test can take several minutes, but with frequencies of 100 kHz - 1 MHz the test time will be too short to be an issue.
