# Strange light polarization effect?

I spent a while working with MgF2-windowed xenon flash / discharge lamps. Primarily, I characterized their spectra with two normal-incidence spectrometers against a calibrated Deuterium lamp. In this particular case, it included a Czerny-Turner-type design.

As side investigation, I rotated one of the xenon flash lamps by 360° in about 45° steps and took time-resolved spectra in the range of 200 to 500 nm for each rotation step. Time-resolved means, that I took measurements / spectral scans at an interval of 0.1 µs with an exposure time of 0.168 µs. Because the exposure time is longer that the interval, I took only one spectral scan per one flash, for each flash delayed by another 0.1 µs. I assume this is ok because the intensity of the light emitted by the discharge over its entire spectrum is stable with $$\sigma = 0.0294$$ So for each rotation step and each interval, I integrate over an entire calibrated spectrum. If I plot the result, it looks somewhat like this:

(The actual flash is "visible" for about 1.5 µs, while there is an "afterglow" mostly in longer wavelengths with less than 5% of the maximum intensity for about another 40 µs.)

My understanding is, that normal-incidence spectrometers behave like polarization filters. So I would expect perfect ellipses - point symmetric - if the the light is (partially) polarized. What I see looks different, unsymmetrical.

If have also done another type of plot. The first one shows how it theoretically should look like:

And this is what I get:

I assume, that I see ellipses. The semi-major-axis / maximal radius is indicated red (1), the semi-minor-axis / minimum radius is indicated blue (-1).

My questions ... is there any physical context, that could explain the unsymmetrical behaviour (other than uncertainties in measurements?). If there is not, would it be appropriate to fit ellipses into the data?

EDIT (1): Based on the reply by @akhmeteli, I looked into different wavelengths. Top left: 230 nm; top right: 260 nm; bottom left: 362 nm; bottom right: 461 nm. The first three represent spectral lines, the fourth one is a random choice without any specific feature. My data has a resolution of 0.2nm. Here, I integrated from lambda-0.2 to lamda+0.2 nm, kind of as narrow as possible. My impression is, that the observed effect does not (very much) depend on the wavelength.

-
I had cause to use a (plain boring Argon-nitrogen) flash lamp once, and have since then had a soft spot for them (actually I have a soft spot for lots of antiquted kit that still find the occasional specialized use). Mind if I ask what your application is? –  dmckee Dec 17 '12 at 14:04
Plain boring ... yep, that's a nice description of my lamps. They were used for photo-ionizing dust / particles, basically. –  ernestopheles Dec 18 '12 at 10:40
I assume the flash lamp is in some kind of transparent enclosure having its own refraction pattern of some unhelpful shape. I wonder if that's contributing to what you see. –  Mike Dunlavey Dec 18 '12 at 14:25